Maintain your equipment to prevent future problems and unwanted costs. Keep your M.O.L.E. Profiler at peak performance by taking care of the system components and keeping it calibrated.
Typical maintenance should include the following:
•Wipe the exterior of the instrument enclosure or components with a cloth dampened with water or IPA alcohol.
IPA alcohol is the only type of solvent that is acceptable to clean Thermal Profiling instruments or any of the system components. All other types such as organic solvents are not recommended.
•Inspect the Power Pack battery charger enclosure for damage. Do not use the charger if the enclosure is cracked or broken, or if the power plug pins are loose or deformed.
•ECD recommends the M.O.L.E. Profiler is factory re-calibrated every 6 months when it is being used constantly. If the use is occasional, a period of no greater than 12 months between calibrations is recommended. The current calibration status can be checked by using the Instrument Status command on the M.O.L.E. menu for all M.O.L.E. Profilers except for the SuperM.O.L.E.® Gold and PTP® VP-8.